LED handpiece integrate E-generator 1 piece. Self-sufficient power system instead of extra supply. Clear view with day-light LED illumination.Operating pressure 0.25Mpa-0.3Mpa(0.3Mpa preferred). M for torque head; 350000r. Speed of generation: 25000 30000 r. Execution according to ISO 7785-1.
Mechanical stability :due to precision of their components, this handpiece should not submitted to a free fall. For dental operational drilling and polishing. Unscrew the cap and take out the cartridge. Put the cartridge into the cave of the head, press it screw the cap with hands first, then use the wrench to fasten it.
Attention: In order to protect the meticulous thread, please do not use the wrench at the beginning. Do not use unqualified bur.
Diamond bur shafts should follow the demands of ISO standard, diameter 1.59-1.60mm. Do not use a bent, damaged, or short hanle bur. Press the cap and take out the bur. Press the cap, insert the bur, and lose the cap. To prevent accident of dropping bur, and damage to beating, bur should be inserted to the bottom.This products is specialized for dental operation. Be careful, this product rotate in high speed when operating.
This product should be used by qualified person. Handpiece is specialized designed for teeth polishing and cutting. Handpiece should be connected with proper tube. This product is specialized for qualified dentist. To prevent medical accident, please read the operation manual before operating.
It is duty of the user to use only faultless tools, to ensure that they are used fot the correct purpose and protect patients and third parties from all dangers. Do not blow spray or blowing air into open wounds during surgery since this may present a risk of air embolism or cutaneous emphyssema. Clean the external side using 70% GL alcohol or other disinfecting without having an ammonium or acid basis or chlorine. Do never immerge the handpiece in disinfection baths.Lubricate the trubine before each sterilization cycle. Handpieces which are not subject to a sterilization process must be lubricated before initiating each work period(in the morning, at noon, and evening). Apply only original lubricants from NSK. An improper care and maintenance may lead to premature wear and failure of the handpiece. Apply only original NSK lubricate spray nozzle. Sterilize the handpiece and bur changer in autoclave up to 135? And 15 minutes with 135?
Withdraw handpiece from autocalve after the sterilization. Clean the LED using 70% GL alcohol or other disinfection without having an ammonium or acid basis or chlorine. Press the O-ring with thumb and index finger, move forward and take out the O-ring. Install new O-ring and spray oil to it. Do not use Vaseline and some other grease.If the handpiece will not be used for a long period it must be cleaned and maintained according to these instructions. Work must stop immediately if there are unnatural running noise, too intense vibrations, excessive eccentricity of the bur or if the bur is not held by the clamp. Before major damage arises consult customer assistance center. Packing and transportation must follow the contract. Relative humidity of storage circumstance should under 80%, no coorrosive and well-ventilated. Cartridge is warranted 90 days, bearing is excluded. Charges resulting from transport of parts of the own product will be the customer. Greatdeshop2019 verpflichtet sich, alle Anforderungen der WEEE-Richtlinie (2002/96 / EG) zu erfüllen. Weitere Informationen zur WEEE-Richtlinie finden Sie unter. Greatdeshop2019 verpflichtet sich auch, die Anforderungen der RoHS-Richtlinie (2002/95 / EG) zu erfüllen oder zu übertreffen. Weitere Informationen zur RoHS-Richtlinie finden Sie unter. The item "10Handstück Dental High Speed LED Fiber Optic Handpiece Turbine 4Loch" is in sale since Wednesday, December 18, 2019.
This item is in the category "Business & Industrie\Medizin & Labor\Zahnmedizin & Kieferorthopädie\Handinstrumente\Allgemeine Handinstrumente". The seller is "greatdeshop2019" and is located in Leipzig. This item can be shipped to all countries in Europe.